Runhua's new intelligent biochemical incubator participated in Chengdu
Information sources： | Release date：
| Browse volume：176
Detailed description of biochemical incubator
Biochemical incubator structure:
The structure of the biochemical incubator adopts high-quality stainless steel inner tank, hollow glass, and polyurethane foam as the insulation material, which has good insulation, light weight, and beautiful overall appearance. The control system adopts microcomputer control, digital display, accurate and intuitive, high precision, convenient and reliable temperature adjustment , There is light for easy observation. The light incubator can be strong or weak; the constant temperature and humidity incubator can also be constant humidity. The refrigeration system is a high-quality fully enclosed compressor with large cooling capacity and low noise. The box is equipped with a high-quality fan device to force air convection, and the temperature distribution is more uniform.
Application of biochemical incubator:
The biochemical incubator is suitable for the cultivation and preservation of bacteria, molds, microorganisms, tissue cells; plant cultivation, breeding experiments; enzymology, enzyme engineering research; biological products, medicines, vaccines, blood and various specimens preservation and testing, it is genetic Ideal laboratory equipment for production and scientific research departments such as engineering, medicine, agriculture, forestry, environmental science, animal husbandry, and aquatic products.
Instructions for use of biochemical incubator:
1. Turn on the power, close the power switch, the whole machine is powered on, and the power indicator light inside the switch is on.
2. For temperature control setting, please refer to the manual of the intelligent digital temperature controller.
3. If lighting is needed, turn on the lighting switch.
4. This incubator has a breakpoint protection function and a one-and-a-half minute delay function. After the compressor stops, it will take about one and a half minutes to restart.
Maintenance and cultivation of biochemical incubator:
1. The shell of the incubator should be grounded reliably.
2. The incubator should be placed in a cool, dry, well-ventilated place away from heat and sunlight. Place it stably to prevent vibration and noise.
3. To ensure effective heat dissipation of the condenser, the distance between the condenser and the wall should be greater than 100mm. There should be a 50mm gap on the side of the box and at least 30cm on the top of the box.
4. When the incubator is transported, repaired, and maintained, avoid collision, shaking and vibration; the maximum inclination is less than 45 degrees.
5. The instrument does not work suddenly, please check whether the fuse tube (behind the box) is burned out, and check the power supply.
6. When refrigerating the biochemical incubator, it is not advisable to make the difference between the temperature in the incubator and the ambient temperature more than 25 degrees
Welcome to the official website of Runhua Instruments. We are a professional manufacturer of laboratory instruments. Quality and reputation a…
The water bath constant temperature shaker is a biochemical instrument that combines a temperature controllable constant temperature water ba…
Series of biochemical incubators:Features of 150A biochemical incubator: the biochemical incubator has the function of cooling and heating tw…
High-precision stirring water bath produced by Runhua Instruments:1) The general water bath does not have the circulating stirring function. …
Maintenance and maintenance of electric mixer1. Correct use and attention to the maintenance of the instrument to keep it in good working con…
What work should be done in the use of digital display stainless steel electric heating plate? Digital display stainless steel electric h…