Overview：1、 Product introduction: The structure of the new constant temperature and humidity seed germination incubator is made of high-quality cold plate sheet metal die. The inner tank is made of high-quality stainless steel material. The observation window…Consulting
1、 Product introduction:
The structure of the new constant temperature and humidity seed germination incubator is made of high-quality cold plate sheet metal die. The inner tank is made of high-quality stainless steel material. The observation window glass door is made of high-quality thermal insulation materials. The overall appearance is beautiful. The control system adopts LED microcomputer control and digital display, which is accurate and intuitive, high-precision, convenient and reliable temperature regulation, and convenient for observation. The refrigeration system is a high-power and high-quality compressor with high speed and low noise. The box is equipped with 9 high-quality bearing elastic fans to force air convection and make the temperature distribution more uniform. The purpose of the seed incubator is: it is suitable for the cultivation and preservation of bacteria, mold, microorganism, tissue cells, plant cultivation, breeding test, mycology, mold engineering research; the preservation and test of biological products, drugs, vaccine blood and various specimens; it is the production of genetics, medicine, agriculture, forestry, environmental science, animal husbandry, aquatic products, etc And scientific research department of the ideal experimental equipment.
2、 Technical parameters:
1. Effective volume: 2841l
2. Temperature control range: 5-50 ℃
3. Temperature control accuracy: ± 0.5
4. Temperature fluctuation: ± 0.1
5. Heating power: 2250w
6. Power supply voltage: 220V50Hz
7. Working size: 1850 x 1280 x 1200 mm
8. Each instrument is equipped with 36 pallets
9. Cooling power: 900W
10. Overall dimension: 1960 x 1580 x 1880 mm
11. R134a environmental protection and fluorine-free refrigeration is used as refrigerant.
12. Humidity range: 40-95% [the humidity content can be determined according to the duration of humidification]
13. Humidity control mode: humidity switch
3、 Instructions for use:
1. Turn on the power, close the power switch, the whole machine is powered on, and the power indicator in the switch is on.
2. Please refer to the manual of intelligent digital temperature controller for temperature control setting. Keep dry and ventilated. All switches on the front panel should be in non working state. The experimental samples placed in the incubator should be kept at a proper distance. It is conducive to the convective circulation of cold and hot air. Turn on the power supply, turn on the switch, and the indicator light will be on; if the cultivation temperature is selected and humidity is required, turn on the humidity switch.
3. If lighting is needed, turn on the lighting switch and set the temperature. Press the set key, and the data in the lower row will flash, which means entering the setting state. Press the arrow down key and the set value will decrease. After pressing the set key to the required setting temperature, it means that the working state is entered. The temperature setting is completed and the setting value is increased. Pressing the arrow up key is the same. [see the instrument operation manual for detailed operation instructions]
4. The incubator has a power-off protection function and a minute and a half delay function. The compressor will start again for about one minute after shutdown.
4、 Maintenance of biochemical incubator:
1. The shell of incubator should be grounded reliably
2. The incubator should be placed in a cool, dry, well ventilated place away from heat sources and sunlight, and placed stably to prevent vibration and noise.
3. In order to ensure the effective heat dissipation of the condenser, the distance between the condenser and the wall should be greater than 10 cm, the side of the box should have a space of 5 cm, and the top of the box should have a space of 30 cm.
4. The incubator should avoid collision, shaking and vibration during handling, repair and maintenance, and the maximum gradient should be less than 45 degrees.
5. If the instrument doesn't work suddenly, please check whether the fuse (behind the box) is burnt out and check the power supply.
Changzhou Runhua Electric Appliance Co., Ltd
Address: No.26, Jinsheng East Road, Jintan City, Jiangsu Province
Tel: 0519-82305398 82301086
Welcome to the official website of Runhua Instruments. We are a professional manufacturer of laboratory instruments. Quality and reputation a…
The water bath constant temperature shaker is a biochemical instrument that combines a temperature controllable constant temperature water ba…
Series of biochemical incubators:Features of 150A biochemical incubator: the biochemical incubator has the function of cooling and heating tw…
High-precision stirring water bath produced by Runhua Instruments:1) The general water bath does not have the circulating stirring function. …
Maintenance and maintenance of electric mixer1. Correct use and attention to the maintenance of the instrument to keep it in good working con…
What work should be done in the use of digital display stainless steel electric heating plate? Digital display stainless steel electric h…