Overview：Hh-4j 4-hole digital display constant temperature stirring circulating water bathDigital display constant temperature stirring circulating water bath pot is widely used in drying, concentration, distillation, impregnation of chemical reagents, drugs a…Consulting
Hh-4j 4-hole digital display constant temperature stirring circulating water bath
Digital display constant temperature stirring circulating water bath pot is widely used in drying, concentration, distillation, impregnation of chemical reagents, drugs and biological products, and can also be used for water bath constant temperature heating and other temperature tests. It is an essential tool for biology, genetics, aquatic products, environmental protection, medicine, health science, biochemistry laboratory, education and scientific research.
The main features of 4-hole digital display stirring water bath are as follows:
1. The working room water tank is made of high-quality stainless steel, which has superior corrosion resistance.
2. Intelligent temperature control, accurate temperature control, digital display and automatic temperature control.
3. In order to ensure the accuracy, advanced electronic technology is adopted to improve the temperature control accuracy.
4. Easy to operate and safe to use.
5. Magnetic stirring function
6. Digital display constant temperature stirring circulating water bath
Temperature range: room temperature ～ 100 ℃
Temperature control digital display automatic temperature control
Temperature uniformity ± 0.5 ℃
Resolution 0.1 ℃
Temperature control range: room temperature ~ 100 ℃
Volume (mm) 320 × 320 × 250
Power supply AC 220 V 50 Hz
Stirring function: 4 holes × 4 holes magnetic stirring
The mixing quantity can be customized according to the needs of users
Operation and maintenance of mixing water bath pot:
1. The constant temperature stirring water bath pot should be placed flat on the fixed platform. The power supply voltage must be consistent with the voltage required by the box. The power socket should be three hole safety socket, and the ground wire must be properly connected before use.
2. Before the experiment, according to the experimental requirements, add appropriate amount of clean water or purified water into the water tank, to the partition board or higher. Then put the experimental vessel on the shelf and cover the hole.
1. Clean the instrument immediately after each use. Put the water in the water bath pan clean, brush off the coarse impurities in the water bath pot with a brush, and clean the pot. Wipe the inside and outside surface of the water bath pan with a soft cloth, and then dry it with a cleaning cloth.
2. There should be no water droplets on the inner surface of the equipment, and the outer surface of the equipment should be bright and clean without stains.
3. When using the water bath pan, it must be grounded reliably, and the water should not overflow into the control box.
4. Do not have water in the water bath pot or the water level is lower than the electric heating tube, in order to prevent the explosion damage of the electric heating tube.
5. Carry out the maintenance and inspection of the instrument once a month, and fill in the maintenance record in time.
Welcome to the official website of Runhua Instruments. We are a professional manufacturer of laboratory instruments. Quality and reputation a…
The water bath constant temperature shaker is a biochemical instrument that combines a temperature controllable constant temperature water ba…
Series of biochemical incubators:Features of 150A biochemical incubator: the biochemical incubator has the function of cooling and heating tw…
High-precision stirring water bath produced by Runhua Instruments:1) The general water bath does not have the circulating stirring function. …
Maintenance and maintenance of electric mixer1. Correct use and attention to the maintenance of the instrument to keep it in good working con…
What work should be done in the use of digital display stainless steel electric heating plate? Digital display stainless steel electric h…