Overview：1、 Product introduction: New drug stability test box The structure is made of high-quality cold plate integral sheet metal. The inner tank is made of high-quality stainless steel liner and insulating glass. The insulation material is polyurethane foa…Consulting
1、 Product introduction:
New drug stability test box The structure is made of high-quality cold plate integral sheet metal. The inner tank is made of high-quality stainless steel liner and insulating glass. The insulation material is polyurethane foam. It has good heat preservation, light weight and beautiful overall appearance. The control system adopts microcomputer LCD control, intelligent program control and digital display. The experimental data is printed by microcomputer, which is accurate, intuitive, high-precision, and convenient for observation. The refrigeration system is a high quality hermetic compressor. With high cooling speed and low noise, the box is equipped with three high-quality fans, forced air convection, more uniform temperature distribution and soft light. It is suitable for the cultivation and preservation of bacteria, mold, microorganism, tissue cells, plant cultivation, breeding test, mycology and mildew engineering research; the preservation and test of biological products, drugs, vaccine blood and various specimens is genetic and medical It is an ideal experimental equipment for agricultural, forestry, environmental science, animal husbandry, aquaculture and other production and scientific research departments.
2、 Technical parameters:
1. Effective volume: 500L
2. Temperature control range: 0-60 ℃
3. Temperature control accuracy: ± 1-2 ℃
4. Temperature fluctuation: ± 0.5
5. Heating power: 600W
6. Power supply voltage: 220V50Hz
7. Working size: 720x600x1200mm
8. Tray for instrument: 3 pieces [pallet can be added or customized according to user's requirements]
9. Cooling power R134a (when the incubator is cooled, it means that when the temperature of the incubator is 16-28 ℃, and the box is over 30 ℃, the compressor is easy to heat up and the automatic protection will be suspended)
10. Humidity range: 40-95% [the humidity content can be determined according to the duration of humidification] humidity accuracy: ± 3-8 ℃
11. Timing range: 1-9999 min
3、 Instructions for use:
1. Turn on the power, close the power switch, the whole machine is powered on, and the power indicator in the switch is on.
2. Please refer to the manual of intelligent digital temperature controller for temperature control setting. Keep dry and ventilated. All switches on the front panel should be in non working state. The experimental samples placed in the incubator should be kept at a proper distance. It is conducive to the convection circulation of cold and hot air. Turn on the power supply, turn on the switch, and the indicator light is on; select the culture temperature, set the temperature you need, and then you can run it. When the constant temperature is stable, the humidity will automatically turn on the humidification.
Special attention: when humidification is needed, the temperature in the chamber can not be designed to zero, otherwise it will be frozen and cannot be humidified.
3. If lighting is needed, turn on the lighting switch and set the temperature. Press the set key, and the data in the lower row will flash, which means entering the setting state. Press the arrow down key and the set value will decrease. After pressing the set key to the required setting temperature, it means that the working state is entered. The temperature setting is completed and the setting value is increased. Pressing the arrow up key is the same. [see the instrument operation manual for detailed operation instructions]
4. The incubator has the function of power-off protection and a minute and a half delay function. It takes about five minutes for the compressor to start again after shutdown.
4、 Maintenance of incubator:
1. The shell of incubator should be grounded reliably
2. The incubator should be placed in a cool, dry, well ventilated place away from heat sources and sunlight, and placed stably to prevent vibration and noise.
3. In order to ensure the effective heat dissipation of the condenser, the distance between the condenser and the wall should be greater than 10 cm, the side of the box should have a space of 5 cm, and the top of the box should have a space of 30 cm.
4. The incubator should avoid collision, shaking and vibration during handling, repair and maintenance, and the maximum gradient should be less than 45 degrees.
5. If the instrument doesn't work suddenly, please check whether the fuse (behind the box) is burnt out and check the power supply. 6. The difference between the temperature in the incubator and the ambient temperature should not be greater than 25 ℃
Welcome to the official website of Runhua Instruments. We are a professional manufacturer of laboratory instruments. Quality and reputation a…
The water bath constant temperature shaker is a biochemical instrument that combines a temperature controllable constant temperature water ba…
Series of biochemical incubators:Features of 150A biochemical incubator: the biochemical incubator has the function of cooling and heating tw…
High-precision stirring water bath produced by Runhua Instruments:1) The general water bath does not have the circulating stirring function. …
Maintenance and maintenance of electric mixer1. Correct use and attention to the maintenance of the instrument to keep it in good working con…
What work should be done in the use of digital display stainless steel electric heating plate? Digital display stainless steel electric h…