Overview：1、 Introduction to oscillating incubator: Constant temperature oscillator (also known as constant temperature oscillation shaker) is used in bacterial culture, fermentation, hybridization, biochemical reaction, enzyme and cell tissue research, etc. i…Consulting
1、 Introduction to oscillating incubator:
Constant temperature oscillator (also known as constant temperature oscillation shaker) is used in bacterial culture, fermentation, hybridization, biochemical reaction, enzyme and cell tissue research, etc. it can be used for the dynamic and static culture of microbial cells and all kinds of bacteria. Set constant temperature culture and oscillation in one, one machine dual-purpose. It adopts high quality compressor and circulating fan, environmental protection, fluorine-free refrigeration, DC brushless motor, long life, wide speed regulation, constant torque, constant speed, maintenance free. Concept aesthetic design with observation window. The inner tank and vibration table are made of anti bacteria stainless steel, which is easy to clean. The unique speed control circuit ensures that the shaking table starts slowly and accelerates steadily. It can prevent liquid splashing and ensure the safety of experimental samples.
2、 Characteristics of oscillation incubator:
1 the insulation material of the box is polyurethane foam, which has strong anti-interference ability to the external cold (hot) source.
02. Air duct is set in the working chamber, and the temperature distribution is uniform
03. The inner wall is made of mirror stainless steel with good corrosion resistance
04. The universal spring bottle rack is especially suitable for the culture and preparation of biological samples for various comparative tests
05. The constant speed meter can set the oscillation speed arbitrarily
06. Accurate temperature control, LCD display
07. There are lighting devices in the box
08. Single gate double layer oscillation
3、 Technical parameters
1. Main application: it is used for multi-purpose constant temperature oscillation culture of samples.
2. Working condition: working voltage: 220 V, 50 Hz, using capacity: 133 L
3. Working environment: working temperature: 15-35 ℃; relative humidity: < 65%.
4、 Technical indicators:
1. The automatic control makes the temperature and speed reach the set value quickly, and ensures the stability of these parameters in the operation process.
2. The LCD display clearly reflects the setting parameters and operation status.
3. Rotation frequency: 0-300rpm
4. Frequency accuracy: ≤± 1 rpm
5. Shimmy amplitude: ￠ 25mm
6. Tray size: 400 × 340mmx2 layers
7. Number of pallets: 2 pieces
8. Temperature control range: normal temperature + 5 ℃ - 65 ℃
9. Temperature control accuracy: ≤ 0.1 ℃
10. Uniformity: ≤± 0.5 ℃
11. Digital display mode: LCD display data
12. Working size: 470 × 400 × 710mm;
13. Weight: 75 kg
5、 Instructions for use of oscillating incubator:
1. The instrument should be placed on the flat ground, the environment should be clean and tidy, dry and ventilated.
2. Before using the instrument, all control switches should be in non working state, and the speed control knob should be at the minimum position.
3. In order to ensure good balance performance of the instrument during oscillation and avoid large vibration, all bottle positions should be filled during bottling, and the culture medium of each bottle should be roughly equal. If the culture medium is insufficient, the test bottle can be placed symmetrically.
4. Turn on the external power supply, turn the power switch to the "on" position, and the indicator light will be on.
5. Select the culture temperature: please refer to the instruction manual.
6. Turn on the "constant speed digital display controller", the blue key, the indicator light is on, slowly adjust the speed control knob to increase the required speed. The service life of the instrument can be prolonged if it is used at medium speed within the speed range.
7. Before each shutdown, all control switches should be in non working state and cut off the power supply
Friendly reminder: the internal parameters of the liquid crystal digital display meter of the oscillation incubator have been set up in the normal working procedure before leaving the factory. Generally, there is no need to change the internal parameters without special requirements, so as to avoid work errors. Please operate according to the instrument operation manual when using, thank you!
6、 Maintenance of oscillation incubator:
1. Correct use and attention to the maintenance of the instrument, make it in a good working state, can extend the service life of the instrument.
2. After the refrigeration system stops working, clean the working chamber and glass observation window with a soft cloth.
3. During the continuous operation of the instrument, regular inspection should be carried out every three months; check whether there are water drops and dirt falling on the motor and exposed refrigeration components; clean the dust and dirt on the compressor and condenser; check the fuse, control element and fastening screw.
The natural wear of the oscillating incubator is normal after long-term use. After the full temperature oscillation incubator is free of warranty, our company will continue to provide quality service and lifelong maintenance.
The above constant temperature oscillation incubator model can not meet your requirements, we can produce customized according to the requirements.
Welcome to the official website of Runhua Instruments. We are a professional manufacturer of laboratory instruments. Quality and reputation a…
The water bath constant temperature shaker is a biochemical instrument that combines a temperature controllable constant temperature water ba…
Series of biochemical incubators:Features of 150A biochemical incubator: the biochemical incubator has the function of cooling and heating tw…
High-precision stirring water bath produced by Runhua Instruments:1) The general water bath does not have the circulating stirring function. …
Maintenance and maintenance of electric mixer1. Correct use and attention to the maintenance of the instrument to keep it in good working con…
What work should be done in the use of digital display stainless steel electric heating plate? Digital display stainless steel electric h…