Overview：1、 Brief introduction of the instrument Temperature controlled sample Cooler: it is widely used and has great effect. It is used for cooling silicon steel bucket samples in iron and steel enterprises. It filters iron oxide slag and iron sheet. The in…Consulting
1、 Brief introduction of the instrument
Temperature controlled sample Cooler: it is widely used and has great effect. It is used for cooling silicon steel bucket samples in iron and steel enterprises. It filters iron oxide slag and iron sheet. The inside and outside of the cooler are made of high-quality thickened stainless steel plate, which is durable and easy to handle. Built in high-quality and durable heating pipe, and intelligent function to prevent dry burning. The box has three layers of protection, shell, inner tank and heat preservation. Equipped with 304 stainless steel filter cage, durable, equipped with inlet valve, overflow valve and water outlet, to meet the needs of users, improve work efficiency, is the ideal equipment for iron and steel production enterprises.
2、 Instrument parameters:
1. Heating power (kw) 8000w
2. Working volume (mm) 760 * 400 * 500 filter tank volume: 400x300x300mm filter cage aperture: 1mm
3. Working power supply (V) 380V, 50Hz
4. Constant temperature range (℃) room temperature - 100 ℃ arbitrary control, high temperature control accuracy
5. Temperature control accuracy: ± 0.1-0.5 ℃ (if water circulation function is needed, please inform our sales department before purchasing, and the cost will be added)
6. With inlet valve, overflow valve and outlet valve
7. Equipped with insulation cover
8. With automatic power-off, anti dry burning, leakage protection function.
1. When using, water must be added into the pot first, and then the power supply is connected. To set the temperature, press the set key to enter the temperature setting mode to set or view the set value. Press the set key to see the number flashing. In the setting mode, press the up button to increase the setting value and the down button to decrease the setting value. If you keep pressing the button, it will increase or decrease to the value you need. After setting, press the set button again to exit the setting mode and return to the normal working measurement mode.
Please note: the following parameters have been adjusted by our company technically. There are no special requirements. Try not to change the parameters that have been set before delivery, so as to avoid other faults. Thank you.
2. Adjust parameter SC: press the set key for 3 seconds and enter the internal parameter setting adjustment interface. The interface shows that SC is the error calibration value, indicating the error between the measured value and the actual value of the temperature controller. The default value is 0.0. You can press the set key again to display the flashing number to enter the adjustment state (the following parameter setting method is the same), and press the up or down button to adjust the value. （-9.9~9.0）
3. E: after setting, press set to enter the next parameter adjustment. The interface shows that e represents the advance of breakpoint control (only effective when p value is set to 0), and the default value is 0.5. The adjustment method is the same as SC.
4. P: press the set key again to enter the third parameter. The interface shows that P represents the setting of P value, and the meaning of P value is roughly the starting control temperature = set value - (10 / P value). If it is set to 50 degrees, when p is 10, it means that 49 degrees begin to control; when p is 4, it means that adjustment begins at 47.5 degrees. If the temperature overshoot is too large, adjust the p value appropriately. If the temperature rises too slowly or the control temperature is not reached, adjust the p value appropriately.
5. Note: when p value is 0, the breakpoint control mode is entered. That is, the system PID fails. When the temperature is lower than (set value - E value) , it will start heating, otherwise it will stop heating.
6. C: press the set key again to enter the setting of parameter C. the parameter is reserved in the system temporarily. Please do not adjust it. The default value is 20.0. After setting all parameters, press set key to return to normal working state. Do not add too much water in the pot, so as to avoid water overflowing out of the boiler when boiling. Pay attention to that the water in the pot can not make the electric heating pipe out of the water surface, so as to avoid burning the electric heating pipe and causing water leakage.
4、 Note: friendly reminder
When the water tank is heated, the water temperature is higher and lower. During the heating process, you can use the foam board or other planks to move the water back and forth to achieve the high precision temperature control function. When you need to use the thermometer to measure, you must measure the water temperature when it is stable. Stainless steel metal sensor, do not collide and move, in order to avoid temperature control failure.
Made by Changzhou Runhua Electric Appliance Co., Ltd
Address: 26 Jinsheng East Road, Jintan City, Jiangsu Province
Welcome to the official website of Runhua Instruments. We are a professional manufacturer of laboratory instruments. Quality and reputation a…
The water bath constant temperature shaker is a biochemical instrument that combines a temperature controllable constant temperature water ba…
Series of biochemical incubators:Features of 150A biochemical incubator: the biochemical incubator has the function of cooling and heating tw…
High-precision stirring water bath produced by Runhua Instruments:1) The general water bath does not have the circulating stirring function. …
Maintenance and maintenance of electric mixer1. Correct use and attention to the maintenance of the instrument to keep it in good working con…
What work should be done in the use of digital display stainless steel electric heating plate? Digital display stainless steel electric h…