Overview：1、 Brief introduction of the instrument Hh-w digital display constant temperature water tank is a comprehensive product of water bath pot, water temperature box and boiling disinfection box. It is suitable for biochemical experiments of distillation,…Consulting
1、 Brief introduction of the instrument
Hh-w digital display constant temperature water tank is a comprehensive product of water bath pot, water temperature box and boiling disinfection box. It is suitable for biochemical experiments of distillation, drying, concentration and constant temperature heating of chemical drugs, biological products and serum examination. It can be used for boiling disinfection of syringes and small surgical instruments. The instrument has the characteristics of excellent material selection, reasonable structure, beautiful appearance, economic, practical, easy operation, convenient maintenance and so on. It is an ideal test equipment.
2、 Structural features:
The shell of digital display constant temperature water tank adopts high-quality thickened national standard cold plate, high-temperature electrostatic spraying, and equipped with stainless steel insulation cover. The heating mode is U-type immersion stainless steel heating pipe, and the temperature control adopts digital LED display. The reading is intuitive and accurate. It can be adjusted and used at will according to the experimental temperature requirements within the experimental range.
3、 Technical parameters:
1. Heating power: 3000W
2. Working volume: 650 * 650 * 400mm
3. Working power supply: 220 V 50 Hz ± 10%
4. Temperature fluctuation: ± 0.5 ℃
5. Temperature control range: normal temperature - 100 ℃
6. New flat cover
7. Uniformity: ± 1 ℃
8. Water tank material: 304 stainless steel
9. Instrument model: hh-w
When using, water must be added into the pot first, and then the power supply is connected. To set the temperature, press the set key to enter the temperature setting mode to set or view the set value. Press the set key to see the number flashing. In the setting mode, press the up button to increase the setting value and the down button to decrease the setting value. If you keep pressing the button, it will increase or decrease to the value you need. After setting, press the set button again to exit the setting mode and return to the normal working measurement mode. After setting all parameters, press set key to return to normal working state. Do not add too much water in the pot, so as to avoid water overflowing out of the boiler when boiling. Pay attention to that the water in the pot can not make the electric heating pipe out of the water surface, so as to avoid burning the electric heating pipe and causing water leakage.
This instrument adopts three pin safety plug. Please use the safe three hole socket and properly grounded. Since the water temperature of the water tank is relatively high when heating, when you need to use a thermometer to measure, you must wait until the water temperature is stable. There is a stainless steel metal sensor in the middle of the heating pipe. Do not collide and move it to avoid temperature control failure.
Changzhou Runhua Electric Appliance Co., Ltd
Address: 26 Jintan Road, Jiangsu Province
Tel: 0519-82305398 82301086
Welcome to the official website of Runhua Instruments. We are a professional manufacturer of laboratory instruments. Quality and reputation a…
The water bath constant temperature shaker is a biochemical instrument that combines a temperature controllable constant temperature water ba…
Series of biochemical incubators:Features of 150A biochemical incubator: the biochemical incubator has the function of cooling and heating tw…
High-precision stirring water bath produced by Runhua Instruments:1) The general water bath does not have the circulating stirring function. …
Maintenance and maintenance of electric mixer1. Correct use and attention to the maintenance of the instrument to keep it in good working con…
What work should be done in the use of digital display stainless steel electric heating plate? Digital display stainless steel electric h…